Water for COD - Chemical Oxygen Demand
Water is used in COD tests for blank preparation, sample dilutions and glassware rinsing. The following water contaminants may have effects on the COD test:
Since the COD test measures the oxygen demand of organic compounds in a sample of water, it is important that no outside organic material be accidentally added to the sample to be measured, as it may give falsely high results.
It is best to minimize their levels in the dilution water, as they may release organics.
Oxidizable inorganic materials may interfere with the determination of COD. They may be oxidized by dichromate and give erroneously high COD results: chlorides are often the most serious source of interference. Nitrites, sulfides and disulfides, sulfites, thiosulfates and ferrous ions may also interfere with the COD test.
Distilled water has been used to prepare dilution water; however, chlorine may co-distil with water and interfere with the COD test. Therefore, distilled water would require an additional sodium thiosulfate treatment. Distillation from alkaline permanganate was sometimes recommended, but this purification procedure is quite cumbersome as well.
Deionized water, purified only by ion exchange resins, may contain organics and microorganisms, thereby causing high COD results. Since the level of organic contamination of deionized water may vary with time, it is not recommended for this test.
Water purified with a combination of technologies, such as reverse-osmosis, ion exchange and/or electrodeionization and activated carbon is low in organics and oxidizable inorganic substances. It is therefore best fitted for the COD test.
Feel free to express your commentsYour opinion is important for us. Please let us know if the information provided was useful, if there is missing information that you would like to see or if you have experience that you would like to share with us on this topic. Please contact us by e-mail, send us feed back and we will come back to you. Contact us.